I’m not dead!
Some time ago I scored a great deal on eBay for a Data I/O 29A setup with a load of modules and manuals.
Pretty much everything worked great apart from the display was really dim.
It doesn’t look too bad in the picture but it looked much worse in real life
Schematics are readily available for the unit so fault finding was fairly quick.
With the display being dim my initial though was a voltage issue and the schematics have conveniently labelled the area up as “FILAMENT SUPPLY GENERATOR”.
Using the scope I could see straight away the CD4049 hex inverter at U14 was not inverting at all.
Simply replacing this fixed the fault